G01NET-4C data acquisition instrument is a dynamic data acquisition instrument with the functions of synchronous acquisition of monitoring data, on-site storage and real-time data upload.This acquisition instrument has 8 channels of analog input channels; it mainly collects low-frequency signals such as dynamic inclinometer, dynamic strain gauge, low-frequency displacement gauge; supports the host computer network to calibrate the acquisition instrument
This instrument supports customers' secondary development. We can open the communication protocol of the collector for free.
The instrument is equipped with special parameter setting software, data acquisition and analysis software. For detailed use of the software, please read the software instruction manual (G01NETCS2019S1 data acquisition instrument parameter setting software instruction manual, G01NETFDC2020S1 wind power tower safety monitoring system instruction manual, DZJC2020S1 project ground vibration and vibration monitoring system instruction manual)
2.The main technical parameter
Number of input channels
Input range and resolution
1~20Hz（All channels are fully synchronized）
Storage capacity and format
Storage capacity: Standard 16G SD card, data storage format: text format
Remote data transmission
Support three communication methods: UDP, TCPIP Client and TCPIP Server; the default is TCPIP Server; the default is continuous transmission after power on
Time correction function
Manual calibration, remote server calibration
4.3 inch LCD with touch
The power supply
The input impedance
Trigger cutoff frequency
Soft trigger, default 1Hz, settable
On-site data storage trigger mode
Trigger based on amplitude
It is mainly used for long-term monitoring of dynamic tilt, dynamic strain and displacement of high-rise buildings such as fan towers, booster stations and buildings.
4. Product features
High precision, low power consumption, small size, real-time data collection, triggered storage of data, real-time data transmission, touch screen display.
5. The appearance introductionof Instrument
Mark (+) on the connection terminal, connect the sensor's power cord to supply power to the sensor, multiple sensors can be shared.
Mark (-) on the terminal, connect the sensor ground wire and shield wire and temperature compensation wire, multiple sensors can be shared.
The terminals labeled 1~8 belong to the signal channel, connect the sensor signal line。
Power on for 5 seconds, restore the factory Settings of the network port IP and port number of the acquisition instrument
Universal network port, data transmission interface, connecting switch or server
One USB interface, used to connect mouse or U disk.
12V dc input port, power supply to the acquisition instrument, factory standard with power adapter
Terminal type 12V DC input interface
Display the operation status and parameter setting of the acquisition instrument.